The software has detected a Validation Thermocouple temperature of 5°C less than the chamber temperature during the sterilizing phase of the cycle.
Check for a clogged venturi in the left rear of the cassette tray. Clean as needed. End user repair
Check the solenoid for debris and make sure the plunger is not sticking. Field Service Tech. repair
Calibrate Validation thermocouple. Field Service Tech. repair
Please sign in to leave a comment.