The software has detected the Validation thermocouple temperature to be 5°C greater than the chamber during the sterilizing phase of a cycle.
Check for visible steam leaks from the cassette, if the cassette is leaking repair as needed. Replace cassette seal, lid or tray. End user repair
Check for kinked exhaust tubing. End user repair
Check for sticking solenoid. Field Service Tech. repair
Calibrate Validation thermocouple. Field Service Tech. repair